Note that the primaryaddress range is from 0 to 30 but any secondary address must be specified in the address rangeof 96 to Show 1 Likes 1. Clear the WTG bit in the Status Operation Condition register after both transient and acquire triggersequences have completed. Page 23 Programming the DC Source – 3Making MeasurementsThe dc source has the ability to make several types of voltage or current measurements. Pins 1 and 2 are output pins controlledby bits 0 and 1. Then the Configure Measurement VI is called to set the instrument up for the desired measurement.
|Date Added:||2 January 2008|
|File Size:||59.55 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Be sure you are familiar with theinformation on this page before programming the dc source for operation from a controller. Or point us to the URL where the manual is located.
Agilent B Labview programming | Keysight Community
Language Dictionary – 4Measurement CommandsMeasurement commands consist of measure and sense commands. Page 73 Language Dictionary – 4Trigger CommandsTrigger commands consist of trigger and initiate commands.
January 22, January 23, When the interrupt occurs, your labvkew can then act on the event in theappropriate fashion. For other instrument driver types, the examples will be available from the Application Examples subpalette from the corresponding instrument driver.
It then generates agilwnt histogram of the pulsewaveform using bins between the maximum and minimum data points.
This example shows how to configure a measurement and take multiple readings. Pins 1 and 2 are output pins controlledby bits 0 and 1. Make sure that for each successful call to the Initialize VI, you use a matching Close VI to avoid maintaining unnecessary memory resources. Page 41 Language Dictionary – 4Table Revised editions are identified by a newprinting date. Errornumbers are returned in two ways: Once the measurement is complete, the Close VI is called which performs an instrument error query and terminates the software connection to the instrument.
The Status VIs obtain the current status of the instrument or the status of pending operations. The relay is controlled independentlyof the output state. You cannot initiate measurement triggers continuously.
What’s missing? Tell us about it.
Options are identified bynumber A 0 indicates no options are installed. Rate this document Select a Rating 1 – Poor 2 3 4 5 – Excellent. This query requests the dc source to identify any options that are installed. Manual UpdatesThe following updates have been made to this manual since the December printingindicated on the Printing History page. C – Error MessagesTable C LAN disconnects sporadically on my EA.
Refer to these for all non-SCPI commands for example: However, LabVIEW Plug and Play instrument drivers are distributed with their block diagram source code, so you can customize agilwnt for a specific application.
The fastest way to get up and running with your instrument driver is to open one of the pre-built examples. Simple examples show you how toprogram: WINDowThis command sets the window function that is used in output measurement calculations.
Language Dictionary – 4Trigger CommandsTrigger commands consist of trigger and initiate commands. Page 72 4 – Language DictionaryTable The message unit may include a parameter after the header. January 10, 5: Trigger commands control the remote triggering of the dc source.
Get a feed of this content Use this view in a tile. Tell us about it.